ABSTRACT In this paper the system for simulation, measurement and processing in graphical user interface implementa- tion is presented. The received signal from the simulation is compared to that of an actual measurement in the time domain. The comparison of simulated, experimental data clearly shows that acoustic wave propaga- tion can be modeled. The feasibility has been demonstrated in an ultrasound transducer setup for material property investigations. The results of simulation are compared to experimental measurements. Results ob- tained fit some much with those found in experiment and show the validity of the used model. The simula- tion tool therefore provides a way to predict the received signal before anything is built. Furthermore, the use of an ultrasonic simulation package allows for the development of the associated electronics to amplify and process the received ultrasonic signals. Such a virtual design and testing procedure not only can save us time and money, but also provide better understanding on design failures and allow us to modify designs more efficiently and economically.
Cite this paper
nullY. Gandole, "Computer Modeling and Simulation of Ultrasonic System for Material Characterization," Modeling and Numerical Simulation of Material Science, Vol. 1 No. 1, 2011, pp. 1-13. doi: 10.4236/mnsms.2011.11001.
 A. A. Berdyev and B. Khemraev, Russian Journal of Physical Chemistry, Vol. 41, 1976, p. 1490.
A. D. Pierce, “An Introduction to Its Physic Principles and Applications,” Woodburg, New York, 1989.
A. Puttmer, P. Hauptmann, R. Lucklum, O. Krause and B. Henning, “SPICE Model for Lossy Piezoceramic Trans-ducers,” IEEE Transactions on Ultrasonics, Ferroelec-trics and Frequency Control, Vol. 44, No. 1, 1997, pp. 60-66. doi:10.1109/58.585191
G. Benny, G. Hayward and R. Chapman, “Beam Profile Measurements and Simulations for Ultrasonic Transducers Operating in Air,” Journal of the Acoustical Society of America, Vol. 107, No. 4, 2000, pp. 2089-2100.
D. Berlincourt, H. A. Krueger and C. Near, “Important Properties of Morgan Electroceramics,” Morgan Electro-ceramics, Technical Publication, TP-226, 2001.
D. E. Gray, “American Institute of Physics Handbook,” 3rd Edition, Mc Graw-Hill, New York, 1972.
D. F. Evans, J. Thomas, J. A. Nadas and M. A. Matesich, Journal of Physics and Chemistry, Vol. 75, No. 11, 1971, pp. 1714-1722. doi:10.1021/j100906a013
D. K. Cheng, “Field and Wave Electromagnetics,” 2nd Edition, Addison-Wesley, Reading, 1989.
G. S. Kino, “Acoustic Waves: Devices, Imaging and Analog Signal Processing,” Prentice-Hall, Englewood Cliffs, 1988.
L. E. Kinsler, A. R. Frey, A. B. Coppens and J. V. Sand-ers, “Fundamentals of Acoustics,” 3rd Edition, Wiley, New York, 1982.
M. Redwood, “Transient Performance of a Piezoelectric- transducer,” Journal of the Acoustical Society of America, Vol. 33, 1961, pp. 527-536. doi:10.1121/1.1908709
M. G. S. Ali, “Analysis of Broadband Piezoelectric Transducers by Discreet Time Model,” Egyptian Journal of Solids, Vol. 23, 2000, pp. 287-295.
M. Hirsekorn, P. P. Delsanto, N. K. Batra and P. Matic, “Modellling and Simulation of Acoustic Wave Propagation in Locally Resonant Sonic Materials,” Ultrasonics, Vol. 42, No. 1-9, 2004, pp. 231-235.
J. Millman and C. Halkias, “Integrated Electronics,” McHill Ltd., Tokyo, 1972, p. 560.
M. H. Rashid, “SPICE for Circuits and Electronics Using PSPICE,” 2nd Edition, Printice Hall of India Pvt, Ltd., New Delhi, 2002.
R. Krimholtz, D. A. Leedom and G. L. Matthei, “New Equivalent Circuits for Elementary Piezoelectric Trans-ducers,” Electronics Letters, Vol. 6, 1970, pp. 398-399.
S. A. Morris and C. G. Hutchens, “Implementation of Mason’s Model on Circuit Analysis Programs,” IEEE Transactions on Ultrasonics, Ferroelectrics and Fre-quency Control, Vol. 33, 1986, pp. 295-298.
Y. B. Gandole, S. P. Yawale and S. S. Yawale, “Simpli-fied Instrumentation for Ultrasonic Measurements,” Electronic Technical Acoustics, Vol. 35, 2005.
J. L. San Emeterio, A. Ramos, P. T. sanz, A. Ruiz and A. Azbaid, “Modeling NDT Piezoelectric Ultrasonic Trans-mitter,” Ultrasonics, Vol. 42, No. 1-9, 2004, pp. 277-281.
J. P. Sferruzza, F. Chavrier, A. Birer and D. Cathignol, “Numerical Simulation of the Electro-Acoustical Response of a Transducer Excited by a Time-Varying Electrical Circuit,” IEEE Transactions on Ultrasonics, Fer-roelectrics and Frequency Control, Vol. 49, No. 2, 2002, pp. 177-183.
S. H. Lee, “Shear Viscosity of Benzene, Toluene, and p-Xylene by Non-Equilibrium Molecular Dynamics Si-mulations,” Bulletin of the Korean Chemical Society, Vol. 25, No. 2, 2004, pp. 321-324.
W. M. Leach, “Controlled-Source Analogous Circuits and SPICE Models for Piezoelectric Transducers,” IEEE Transactions on Ultrasonics, Ferroelectrics and Fre-quency Control, Vol. 41, 1994, pp. 60-66.
W. P. Mason, “Electromechanical Transducers and Wave Filters,” Van Nostrand, New York, 1942.
W. Schaaff, “Numerical Data and Functional Relational-ships in Science and Technology,” In: K. H. Hellwege and A. M. Hellweg, Eds., New Seris Group II: Atomic and Molecular Physics, Vol. 5: Molecular Acoustics, Springer-Verlag, Berlin, 1967.
Weast, C. Robert, “Handbook of Chemistry and Physics,” 45th Edition, Chemical Rubber Co., Cleveland Ohio, 1964, p. E-28.
X. Q. Bao, Y. Bar-Cohen, Z. Chang, B. P. Dolgin, S. Sherrit, D. S. Pal, S. Du and T. Peterson, “Modeling and Computer Simulation of Ultrasonic/Sonic Driller/Corer (USDC),” IEEE Transactions on Ultrasonics, Ferroelec-trics and Frequency Control, Vol. 50, No. 9, 2003, pp. 1147-1160.