The 60Co gamma irradiator, installed in the Center National for Nuclear Sciences and Technology (CNSTN), is used especially for irradiation purposes. It is designed for sterilization of medical devices and foodstuff irradiation. It is an extended source composed by eight pencils of 45 cm length arranged in two levels. Four pencils are mounted around a 7-cm radius circle, upward along the z-axis, whereas, the other four are mounted around a 4-cm radius circle, downward the z-axis  .
Finally, the overall high of the source is equal to 90 cm and the mean radial extension is about 5.5 cm. In a previous work  , we have shown that, the extended source of the CNSTN is equivalent, less than one percent, to a pencil-like gamma source with the same activity and the same height. Using this previous result, we have shown determine the isodose curves in the vicinity of the CNSTN Irradiator.
2. Multipole Expansion
In order to calculate the photon flux and the dose rate in the vicinity of the 60Co gamma source presented below, the author has used the multipole expansion as function of r’/r, where r' is the vector position of an elementary source activity and r is the vector position of the dosimeter relative to the source  . In its work  , the author has calculated the gamma flux rates in 36 points in the vicinity of the source. The distance between the source and the plane of measurement points starts from 40 cm with a 25-cm step along each direction, which forms a square mesh of 25 cm of length. Unfortunately, the author claims that, he has found a good agreement between the relative calculated values and the measurements.
Note that the multipole expansion, is valid only, if the distance between the source and the point of calculation the photon flux is very large compared to the source dimensions (in other word, the point is far away from the source). Given that, in some positions used by this reference, the ratio r'/r can be even bigger than one (r = 40 cm and r' = 45 cm: r’/r = 1.125). Then a multipole expansion as a function of r'/r, which can never converge, is wrong.
In fact, we can use the multipole expansion to calculate the photon flux, but it must be a development in terms of ρ/r which is always very small, ρ being the mean radial extension of the CNSTN source (ρ ≈ 5.5 cm). Secondly, since we have shown that the CNSTN source was equivalent to a pencil like source, it is easier to determine the dose distribution using the straightforward expression than the multipole expansion.
The expression of the dose rate, in the vicinity of the CNSTN Irradiator, obtained in reference  is given by:
where: is a constant.
In the present, we have shown that the multipole expansion as function of r’/r cannot be used to determine the photon flux as well as the dose distribution in the vicinity of the CNSTN irradiator. In fact, if it is necessary to use such
Figure 1. Isodose curves at seven vertical plans (Gy/h).
development, it will be as function of ρ/r, which is always very small. We have also shown that the dose mapping in the vicinity of the CNSTN Irradiator can be determined using a simple expression.
 Jemii, E., Mazouz, M., Ben Fredj, A. and Ghedira, L. (2011) Modeling of the Tunisian 60Co Gamma Irradiator by a Coaxial Equal Height and Equal Activity Single Pencil. diation Physics and Chemistry, 80, 1158-1161.
 Loussaief, A., Trabelsi, A. and Baccari, B. (2006) Extended Gamma Sources Modelling Using Multipole Expansion: Application to the Tunisian Gamma Source Load Planning. Radiation Physics and Chemistry, 75, 463-472.
 Loussaief, A. and Trabelsi, A. (2007) Dose Mapping Using Multipole Moments. Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment, 580, 102-105.
 Jemii, E., Mazouz, M. and Ghedira, L. (2013) Dose Rate Calculation in the Vicinity of the Tunisian Gamma Irradiation. World Journal of Nuclear Science and Technology, 3, 28-32.