Patient safety is defined as the absence of avoidable patient harm during the process of medical attention.
All medical attention brings inherent risk of adverse events (AE) that could cause injury, disabilities and even death of the patient.
Based on studies conducted by Brennan  , in 1999 Kohn published “To err is human: Building a Safer Health System”  . This paper told that at least 1,000,000 of AE happen in the United States yearly, and carry on the death of 44,000 to 98,000 people. This studies a real important revolution in health world, being aware of the error rate attributable to health system that has great impact on patients. At the beginning of the 2000s some initiatives appeared and some strategies were proposed to analyze and to see how you can reduce the rate of preventable errors.
Patient safety is a target for health systems and is a fundamental principle of healthcare, as well as an important component of quality management.
The main global health organizations have incorporated patient safety in their review of work practices. Among these, the World Alliance for Patient Safety, from the 55th Health Conference in October 2004 found that the professional services of health care play an important role in risk management and in creation of safer health systems  . The president of that alliance, L. Donaldson, refers to the role of medical laboratory in patient safety  .
Several studies about risk management and patient safety analyze their involvement in it scope. We highlight the Spanish National Study on Hospitalisation-Related Adverse Events (ENEAS) in 2005  , which was part of a quality program of the national health system and showed that 42.6% of the Adverse Events (AE) were preventable. In the laboratory, AE rate according to some reports by Plebani  fluctuates from 2.7% to 12%.
Nowadays, quality management systems are implemented in medical laboratories. Their aims are reducing potential risks and improve patient safety    .
The information provided by the medical laboratory has a direct impact on patient safety and a fault in any of processes strategic, operational (preanalytical, analytical, postanalytical) and support, could affect patients. An improvement in the safety of the various processes brings to light the potential failure modes in the laboratory and try to solve them.
To provide useful and reliable information to the clinician, it is important to emphasise the need to design risk and processes map in the laboratory  , together with quality indicators that allow monitoring and risk management  .
Our study aims calculate the impact of the failure modes in a medical laboratory and compare the risk with two risk management tools: Failure Mode and Effects Analysis (FMEA) versus the Failure Reporting Analysis and Corrective Action System (FRACAS). Use FMEA to estimate the potential risks and FRACAS to make real errors analysis.
The scope of application is all processes in the medical laboratory (Catlab) at Consorcio Sanitario de Terrassa Hospital (CST). Medical laboratory was certified with ISO 9001:2000 Quality Management since 2004 and nowadays has been accredited according to UNE-EN ISO 15,189:2013.
Consorcio Sanitario de Terrassa hospital manages 340 beds for intensive care patients and 32 beds for penitentiary patients. In its hospital network provides services to 34 primary care centres and two specialized primary care centres. It serves a population of almost 400,000 people.
Process map of medical laboratory was made with Visio Standard 2007 Microsoft Office program (Figure 1), which provided a global perspective of the laboratory.
The study was made about 90 possible modes of failure detected by the Failure Model and Effects Analysis (FMEA) model applied to laboratory processes   . FMEA is a preventive and proactive tool. It analyzes the quality, safety and/or reliability of a system performance operation, identifying possible failure modes presented, and to apply preventive actions to avoid problems that could be manifested themselves in the future.
The failure modes were identified from the literature    and a brainstorming conducted among a working group of laboratory professionals.
FMEA allowed identify potential failure modes and estimate risk through a table of three variables (Table 1): severity, frequency and detection   . The severity score variable is based on a scale from 1 to 10, being 1 the least severe value and 10 the worst. The variable frequency is based on a scale from 1 to 10, being 1 the least likely to appear and 10 the highest. Finally, the variable detection, it is also classified on a scale from 1 to 10, but in this case 10 means a minor probability to detect and 1 a higher one. With the product of these three variables the risk priority number (RPN), has a potential value between 1 and
Figure 1. Processes map of medical laboratory.
Table 1. Risk assessment values.
1000. From this way those risks were evaluated which could have direct or indirect impact on patient safety.
Then Failure Reporting Analysis and Corrective Action System (FRACAS)  was used. It is a corrective and reactive tool that allows analyzing variables associated with the damage and factors that explain it. FRACAS uses only two variables, severity and frequency with the same scale than FMEA for severity. For frequency, annual indicators were calculated and with the objective to harmonize with FMEA  the same scale was used Table 1, they were transformed to the same FMEA frequency scale used. With the product of these two variables the risk priority number (RPN), has a value between 1 and 100.
FRACAS allowed a real calculation of the frequency of detected faults and the severity of them. It is a dynamic tool that can identify and incorporate unanticipated errors in the FMEA  . It was used information from the quality management system of the laboratory (audits, management reviews, indicators, nonconformities, etc.) to do this calculation.
Risk and processes maps were made with a Visio program from the results of FMEA and FRACAS  .
This study allowed the calculation of the potential risk in the preanalytical, analytical and postanalytical processes, as well as strategic and support processes of medical laboratory.
The processes map of medical laboratory shows the activities in each process and a general viewer of laboratory medicine (Figure 1).
The results show the priority risks identified by FMEA. These risks are classified according to the risk priority number (RPN). The five failure modes with maximum NPR for each process are presented in Table 2.
Detected failure modes are classified by FRACAS according to risk priority number (RPN). Five failures with maximum RPN are presented for each process (Table 3).
Table 2. Classified failure modes by FMEA.
Compare the results obtained by FMEA and FRACAS according risk priority. It presents only 10 faults with major RPN according to FMEA (Table 4).
Compare the results obtained by FRACAS and FMEA according the risk priority. It shows only the top10 failures with higher RPN, according to FRACAS (Table 5).
Percentage (%) distribution of failure modes identified according the affected process shows in Table 6.
The results obtained are distributed in the risk map.
The risk map allows us to have a global view on each activity of risk estimation and detection of failure modes. The results show AMFE versus FRACAS in each affected process.
Table 3. Classified failure modes by FRACAS.
Table 4. Comparison of the top risk numbers FMEA versus FRACAS.
Table 5. Comparison of the top risk numbers FRACAS versus FMEA.
Table 6. Failure modes identified (%) by processes according to FMEA and detected faults (%) according to FRACAS.
Any failure in the processes established into the laboratory can lead to consequences in patients, being a key component in relation to patient safety   . That is why we have to manage these failures and implement improvement plans to reduce them  . Nowadays, it is seen a tendency to move from the culture of error detection to the management of risk in all quality systems of clinical laboratories  .
In the literature, authors believe that the study of the impact of risks must be made in operational, strategic and support processes. There are studies showing these processes by designing indicators, such as related to the competence of professionals, customer service  or indicators associated with strategic support processes   or operational and support processes  . It must be stressed, however, that most publications are focused on the operational processes (preanalytical, analytical and postanaytical)     or preanalytical and postanalytical   or exclusively preanalytical   .
This series of quality indicators described in those studies, as well as patient risk, come to meet the need to comply with the strategic lines that are being defined in the health sector, related to the dissemination of the culture of patient safety and the implementation of improvement plans to increase safe practices in this environment.
On the other hand, the two standards of broad application in laboratories (UNE-EN ISO 9001:2015 and UNE-EN ISO 15,189: 2013) are also involved in the risk management of the patient   , although the design of its indicators is not made from the use of tools such as FMEA and FRACAS. Therefore, we found it interesting to carry out this study in all laboratory processes using both FMEA and FRACAS tools because they are widely used in the clinical laboratories to highlight the need for implement risk management.
The application of these tools is not as widespread as indicators of quality of clinical laboratories. However, it is interesting the Astion and colleagues’ study  that analyzed the impact on the patient of incidents in the laboratory and compares real potential adverse events. Another interesting study was done by A. Giménez and colleagues  which used FMEA only in preanalytical processes.
In our study, results from FMEA were obtained with three variables and FRACAS with two variables. Detectability in FRACAS is real because the errors are registered, while FMEA estimates detectability. This fact is reflected in the results presented in Tables 2-5.
Table 6 shows the percentage of distribution of failures in laboratory processes using the FMEA tool and the percentage of distribution of errors using the FRACAS tool. It is noticeable that there is a significant difference in the support processes between FRACAS and FMEA (4.2% FRACAS compared with 1.3% FMEA).
Discrepancies are observed between preanalytical (48.4% FRACAS versus 34.2 FMEA) and postanalytical (17.2% FRACAS versus 35.1% FMEA) processes. However, the results showed a good agreement in analytical processes (28.2% FRACAS versus 26.5% FMEA).
If we compare our results with those obtained by Plebani  , we have similar results for the preanalytical processes (in FRACAS 48.4% versus 46% - 68.2% of Plebani) and the postanalytical processes (35.1% compared FMEA 18.5% - 47% of a Plebani). However, in analytical processes the results do not match the two studies (26.5% FMEA and FRACAS 28.2%, versus 13% and 7% Plebani).
Strategic and support processes contribute to patient risk rate much lower than the operative processes. As regards strategic by low estimated frequency and in support processes due to low gravity failure modes.
The processes map adds information about the organization of processes and subprocesses in clinical laboratory. Together with the risk map, it gives us a global view of the distribution of failures in each of the processes.
It has been made the calculation of RPN, to assess the impact of potential risks. From these results, it could be developed an improvement plan to implement corrective and preventive actions, in accordance with the standards ISO 15,189:2013  and ISO 9001:2015  . Keep in mind that prioritization must be made from the calculation of the failure modes and not from the subprocesses or processes because potentially serious risks (but less frequent) could be masked.
The fact of identifying potential failure modes by FMEA tool makes to review
Figure 2. Risks map of medical laboratory.
meticulously the processes implemented to detect all possible faults in the various activities and stages involved in them. It was decided to make the study of risks in all laboratory processes (operational, strategic and support), since activities performed in all processes can cause potential risks and can have an impact on patient safety as shown in Figure 2.
FMEA allows detecting critical points in terms of the patient risk and FRACAS highlights the priorities to control these points and help to select preventive or corrective actions that we should be incorporated in the laboratory improvement planning.
If FMEA is compared versus FRACAS, the difference is that indices of risk priority are higher in FMEA in postanalytical processes, while comparing FRACAS versus FMEA the rates of risk priority are higher in preanalytical processes.
The greatest impact of potential real errors in patients appear in activities related to operational processes, which are more related to the actions of health professionals on patients.
It is important to note that FMEA is a subjective tool and that to be able to make a real study of failures FRACAS has to be performed.
On the basis of the results obtained of FMEA and FRACAS a strategic risk management plan should be implemented.
It is conclusive the need for risk management in clinical laboratories and monitoring them within the quality plan, a fact that would lead to an increase on patient safety.
 Brennan, T.A., Leape, L.L., Laird, N.M., et al. (1991) Incidence of Adverse Events and Negligence in Hospitalized Patients Results of the Harvard Medical Practice Study I. The New England Journal of Medicine, 324, 370-376. https://doi.org/10.1056/NEJM199102073240604
 World Health Organization (2002) The World Health Report 2002. Reducing Risks, Promoting Healthy Life. WHO, Geneva. http://www.who.int/whr/2002/en/
 Plebani, M., Astion, M., Barth, J., et al. (2014) Harmonization of Quality Indicators in Laboratory Medicine. A Preliminary Consensus. Clinical Chemistry and Laboratory Medicine, 52, 951-958. https://doi.org/10.1515/cclm-2014-0142
 Kenneth Crow DRM Associates. http://www.npd-solutions.com/fmea.html
 Krouwer, J. (2006) Dynamic Rather than Static Performance Measures Are Needed to Improve Patient Safety. Accreditation and Quality Assurance, 11, 644-646.
 Plebani, M., Ceriotti, F., Messeri, G., et al. (2006) Laboratory Network of Excellence: Enhancing Patient Safety and Service Effectiveness. Clinical Chemistry and Laboratory Medicine, 44, 150-160. https://doi.org/10.1515/CCLM.2006.028
 Ricos, C., Biosca, C., Ibarz, M., et al. (2008) Quality Indicators and Specifications for Strategic and Support Processes in Laboratory Medicine. Clinical Chemistry and Laboratory Medicine, 46, 1189-1194. https://doi.org/10.1515/CCLM.2008.225
 Ruiz, R., Llopis, M., Biosca, C., et al. (2010) Indicators and Quality Specifications for Strategic and Support Processes Related to the Clinical Laboratory. Four Years’ Experience. Clinical Chemistry and Laboratory Medicine, 48, 1015-1019. https://doi.org/10.1515/CCLM.2010.229
 Alsina, M., Alvarez, V., Biosca, C., et al. (2007) Quality Indicators and Specifications for Key Processes in Clinical Laboratories; A Preliminary Experience. Clinical Chemistry and Laboratory Medicine, 45, 672-677.
 Laposata, M. and Dighe, A. (2007) Pre-Pre and Post-Post Analytical Error: High-Incidence Patient Safety Hazards Involving the Clinical Laboratory. Clinical Chemistry and Laboratory Medicine, 45, 712-719. https://doi.org/10.1515/CCLM.2007.173
 Astion, M., Shojania, K., Hamill, T., et al. (2003) Classifying Laboratory Incident Reports to Identify Problems That Jeopardize Patient Safety. American Journal of Clinical Pathology, 120, 18-26. https://doi.org/10.1309/8U5D0MA6MFH2FG19
 Giménez, A., Molina, P., Ruiz, J., et al. (2010) Application of Failure Mode and Effects Analysis of the Pre-Analytical Phase in a Clinical Laboratory. Revista del Laboratorio Clínico, 3, 161-170. (In Spanish)