Meter, the unit of length, is defined in vacuum. However, measurements of length are often carried out in air, which presents some problems. Let us assume that we want to compare two geometric distances and. These two distances are measured in air as and, where and are the refractive index of air (RIA). In the absence of a relationship between and, it is not possible to determine which of and is greater only by judging the magnitude relationship between and. To solve this problem, the influence of RIA must be eliminated.
One approach to obtaining the value of RIA is to use empirical equations -. With n obtained, an estimate of the geometric distance can be calculated. The estimated geometric distance can be used for comparison. The empirical equations are used to compensate for the RIA under two assumptions. First, environmental parameters (namely, temperature, pressure, and humidity) can be measured. Second, a measured environmental parameter is a good reproduction of that parameter along the optical path, meaning that a measured environmental parameter is an average value over time and space. In other words, both the spatial distribution of environmental parameters and the time-delay of measurement equipment can be ignored. These assumptions are valid only if the measurement is performed in a closed environment (e.g., a well-controlled laboratory or underground tunnel with limited variation in environmental parameters).
Another approach to suppressing the influence of RIA is to apply the two-color method, which was first proposed by Bender and Owens  to compensate for the inhomogeneous disturbances of the RIA in an open environment. The core concept of the two-color method is to use a measured length difference between two colors (frequencies) to render length measurements less sensitive to changes in the RIA.
Recently, high-precision length measurements based on fem to second optical frequency comb (FOFC) have been carried out (e.g.,  ). To compensate for the RIA, FOFC-based RIA measurements   and FOFC- based two-color method experiments - have also been performed. Minoshima’s group performed a two- color method experiment in a well-controlled environment and found an agreement between RIA compensation based on the empirical equations and that of two-color method with a standard deviation of 3.8 × 10−11 throughout hours . They also suggested that the accuracy provided by the empirical equations may be improved by the two-color method.
One question arises naturally: theoretically, is the two-color method superior to the empirical equations in RIA compensation? We employed a numerical approach to investigate this possibility.
2.1. Refraction Index Compensation by Empirical Equations
The distance between two points measured in air is an optical distance. An estimate of the geometric distance in vacuum and the optical distance has the following relationship.
where n represents the RIA. By applying the law of propagation of uncertainty   to Equation (1), we obtain the uncertainty of length in vacuum.
where denotes the uncertainty of variable x. The first and second terms of the right-hand side of Equation (2) are the uncertainty due to the refractive index and the length measurement, respectively. These two are defined as follows, respectively.
The uncertainty of refractive index can be evaluated by the following equation  .
where, , and are the uncertainties of the instrument for measuring temperature T, barometric pressure P, and humidity H, respectively., , and are sensitivity coefficients and defined as follows.
where is the derivative of function at. The definitions are similar for and.
2.2. Refraction Index Compensation by Two-Color Method
The distances between two points measured in air by using different wavelengths are optical distances and. An estimate of the geometric distance from these two optical distances can be obtained as follows:
where A is the so-called A-factor defined as
Equation (7) can be rewritten as follows.
By applying the law of propagation of uncertainty to Equation (9), we have
The uncertainties of the first and second terms of the right-hand side of Equation (10) are, respectively,
Because we have, Equation (12) can be rewritten as follows.
By substituting Equations (11) and (13) into Equation (10), we obtain
The first and third terms of the right-hand side of Equation (14) are the uncertainty due to the A-factor, and the second and fourth terms are the uncertainty due to the length measurement. These two are defined as follows, respectively.
The uncertainty of A-factor is as follows.
where, , and are the sensitivity coefficients of the A-factor and are defined as follows.
where is the derivative of function at. The definitions are similar for and.
2.3. Comparison of Empirical Equations and Two-Color Method
In Equation (4), the uncertainty due to the length measurement is multiplied by the factor. In Equation (16), the uncertainty due to the length measurement is multiplied by two factors, and. Normally, their orders are several tens. If the two wavelengths used in the two-color method are 780 nm and 1560 nm, then and. By comparing the magnitudes of Equation (4) and Equation (16), we understand that only when the condition
is satisfied, the two-color method can be shown to obtain measurements with a smaller error than that of the empirical equations. We performed numerical calculations to check whether Equation (19) is feasible.
3. Numerical Calculations
We used the following parameters for simulation. By referring to Ref. , we employed 780.0 nm and 1560.0 nm as the two wavelengths. We used the equations for the phase refractive index given in Ref. . Because of the limit on the length of this paper, we only considered the Edlén empirical equations in this study. In the Edlén empirical equations -, the RIA can be derived from the wavelength in vacuum, temperature T, barometric pressure P, and humidity H as. The formula used to perform the calculations can be easily accessed via the internet . In the following, we only consider the phase refractive index. The group refractive index can be treated in the same way.
On the basis of Equations (6) and (18), we calculated the change in the sensitivity coefficients when environmental parameters change in a realistic range (T ∊ [10, 30] ˚C, P ∊ [90,115] kPa, H = 0%). The calculations of the derivative of each refractive index have been validated in Ref. . The same procedure was used in this study for calculating the derivative of the A-factor. After obtaining an expression for the sensitivity coefficients by substituting numerical values, the values of sensitivity coefficients were calculated.
As shown in Figure 1, when = 0%, the sensitivity coefficient of the A-factor is smaller than that of the refractive indices. This result, i.e., the A-factor can be considered as a function of just two wavelengths only when the humidity is 0%, is consistent with the results of previous studies    -.
On the basis of Equations (3) and (15), we calculated the uncertainties due to the A-factor and refractive indices, respectively. The geometric distance G was set to 1 m. We assumed that and on the basis of using a thermometer (Testo 735, Testo) and a barometer (VR-18, Sunoh), respectively. These two are commercially available for us.
Figure 2 shows that. This result means that in a 0% humidity environment, the two- color method has potential to provide greater measurement accuracy than the empirical equations. Note that the orders of values shown in Figure 2 were affected by the sensitivity coefficients of environmental parameters and the uncertainties of the instrument for measuring environmental parameters. A detailed uncertainty analysis in an environment where the humidity is not 0% will be reported in another paper.
We analyzed the uncertainties of length conversion based on the Edlén empirical equations and the two-color
Figure 1. Change in sensitivity coefficients of A-factor and refractive indices with (a) temperature when P = 101.325 kPa and H = 0% and (b) pressure when T = 20˚C and H = 0%.
Figure 2. Change in uncertainties due toA-factor and refractive indices with (a) temperature when P = 101.325 kPa and H = 0% and (b) pressure when T = 20˚C and H = 0%.
method, in which the uncertainties due to length measurement and refractive index compensation were decomposed. Using numerical calculations of sensitivity coefficients of the A-factor and refractive indices of the environmental parameters, we found for the first time that in a realistic environmental parameter range (T ∊ [10, 30] ˚C, P ∊ [90, 115] kPa, H = 0%), the uncertainty of the two-color method due to the A-factor was smaller than that of the empirical equations due to refractive indices. This result suggests that in a 0% humidity environment, the two-color method has potential to provide greater measurement accuracy than the empirical equations, with the cooperation of suppressing the uncertainties of length measurements (compared with uncertainties of refractive index compensation) to a negligible level. The findings of this study provide a better insight into the two- color method, and will create opportunities for further development of application of this method.
This research work was partially financially supported by a grant (R1501) from the Mitutoyo Association for Science and Technology.
 Ciddor, P.E. (1996) Refractive Index of Air: New Equations for the Visible and Near Infrared. Appl. Opt, 35, 1566-1573. http://dx.doi.org/10.1364/ao.35.001566
 Birch, K.P. and Downs, M.J. (1993) An Updated Edlén Equation for the Refractive Index of Air. Metrologia, 30, 155. http://dx.doi.org/10.1088/0026-1394/30/3/004
 Bengt, E. (1966) The Refractive Index of Air. Metrologia, 2, 71. http://dx.doi.org/10.1088/0026-1394/2/2/002
 Stone, J.A. and Zimmerman, J.H. Refractive Index of Air Calculator. http://emtoolbox.nist.gov/Wavelength/Edlen.asp
 Bender, P.L. and Owens, J.C. (1965) Correction of Optical Distance Mea-surements for the Fluctuating Atmospheric Index of Refraction. J. Geophys. Res, 70, 2461-2462. http://dx.doi.org/10.1029/JZ070i010p02461
 Coddington, I., Swann, W.C., Nenadovic, L. and Newbury, N.R. (2009) Rapid and Precise Absolute Distance Measurements at Long Range. Nat Photon, 3, 351-356. http://dx.doi.org/10.1038/nphoton.2009.94
 Lee, J., Kim, Y.-J., Lee, K., Lee, S. and Kim, S.-W. (2010) Time-of-Flight Measurement with Femtosecond Light Pulses. Nat Photon, 4, 716-720. http://dx.doi.org/10.1038/nphoton.2010.175
 Zhang, J., Lu, Z.H. Menegozzi, B. and Wang, L.J. (2006) Application of Frequency Combs in the Measurement of the Refractive Index of Air. Rev. sci. instrum, 77, 083104-083105. http://dx.doi.org/10.1063/1.2239036
 Zhang, J., Lu, Z.H. and Wang, L.J. (2007) Precision Measurement of the Refractive Index of Carbon Dioxide with a Frequency Comb. Opt. Lett, 32, 3212-3214. http://dx.doi.org/10.1364/OL.32.003212
 Minoshima, K. and Matsumoto, H. (2000) High-Accuracy Measurement of 240-m Distance in an Optical Tunnel by use of a Compact Femtosecond Laser. Appl. Opt, 39, 5512-5517. http://dx.doi.org/10.1364/AO.39.005512
 Meiners-Hagen, K. and Abou-Zeid, A. (2008) Refractive Index Determination in Length Measurement by two-Colour Interferometry. Meas. Sci. Technol, 19, 084004. http://dx.doi.org/10.1088/0957-0233/19/8/084004
 Kang, H.J., Chun, B.J., Jang, Y.-S., Kim, Y.-J. and Kim, S.-W. (2015) Real-Time Compensation of the Refractive Index of Air in Distance Measurement. Opt. Express, 23, 26377-26385. http://dx.doi.org/10.1364/OE.23.026377
 Wu, G., Arai, K., Takahashi, M., Inaba, H. and Minoshima, K. (2013) High-Accuracy Correction of Air Refractive Index by Using Two-Color Heterodyne Interferometry of Optical Frequency Combs. Meas. Sci. Technol, 24, 015203. http://dx.doi.org/10.1088/0957-0233/24/1/015203
 Walter, B. Maurice, G.C. and Peter, M.H. (2006) Evolution of the “Guide to the Expression of Uncertainty in Measurement”. Metrologia, 43, S161. http://dx.doi.org/10.1088/0026-1394/43/4/S01
 Ellis, J.D. (2014) Field Guide to Displacement Measuring Interferometry, The Field Guide Series. SPIE, Bellingham. http://dx.doi.org/10.1117/3.1002328
 Wei, D. and Aketagawa, M. (2015) Uncertainty in Length Conversion due to Change of Sensitivity Coefficients of Refractive Index. Opt. Commun, 345, 67-70. http://dx.doi.org/10.1016/j.optcom.2015.01.070
 Ishida, A. (1989) Two-Wavelength Displacement-Measuring Interferometer Using Second-Harmonic Light to Eliminate Air-Turbulence-Induced Errors. Jpn. J. Appl. Phys, 28, L473. http://dx.doi.org/10.1143/jjap.28.l473
 Matsumoto, H. and Honda, T. (1992) High-Accuracy Length-Measuring Interferometer Using the two-Colour Method of Compensating for the Refractive Index of Air. Meas. Sci. Technol, 3, 1084. http://dx.doi.org/10.1088/0957-0233/3/11/011
 Matsumoto, H., Zhu, Y., Iwasaki, S. and Oishi, T. (1992) Measurement of the Changes in Air Refractive Index and Distance by Means of a Two-Color Interferometer. Appl. Opt, 31, 4522-4526. http://dx.doi.org/10.1364/AO.31.004522
 Golubev A.N. and Chekhovsky, A.M. (1994) Three-Color Optical Range Finding. Appl. Opt, 33, 7511-7517. http://dx.doi.org/10.1364/AO.33.007511
 Minoshima, K., Arai, K. and Inaba, H. (2011) High-Accuracy Self-Correction of Re-fractive Index of Air Using Two-Color Interferometry of Optical Frequency Combs. Opt. Express, 19, 26095-26105. http://dx.doi.org/10.1364/OE.19.026095