Pulse X-ray diagnostics is capable of
reducing the radiation exposure considerably. As for pulse X-ray diagnostic machines,
which form pulses with the duration of 0.1 μs, using them one can get
outstanding results in this area. This fact can be explained by the long period
of luminophor persistence in intensifying X-ray luminescent screens. In this
paper we present experimental data, comparing radiation doses, measured at
pulse X-ray apparatus and apparatus of constant radiation.
Cite this paper
Komarskiy, A. , Chepusov, A. , Kuznetsov, V. , Korzhenevskiy, S. , Niculin, S. and Cholakh, S. (2014) Reducing Radiation Dose by Using Pulse X-Ray Apparatus. Journal of Biosciences and Medicines, 2, 17-21. doi: 10.4236/jbm.2014.22003.
 Doroshko, M. V. (1998) New Capabilities in Reducing of Radiation Dose лучевой нагрузки. News of Radiodiagnostics, 3, 13-15.
 Blinov, N.N., Bikov, R.E. and Kozlovskiy, E.B. (1989) Engineering Tools of Medical Introscopy. Leonov B.I., Moskow.
 Kazgikin, O.N., Markovskiy, L.Y., Mironov, I.A., Pskerman, F.M. and Petoshina, L.N. (1975) Inorganic Phosphor. Leningrad.
 Filatov, A.L., Korzhenevskiy, S.R., Kuznetsov, V.L., Ananin, M.V. and Motovilov, V.A. (2004) The Nanosecond PP X-ray Apparatus. Procedings of the 15th International Conference on High-Power Particle Beams, Saint-Petersburg, 1-7 April 2004, 552-554.
 Filatov, A.L., Bastrikov, V.L., Korzhe-nevskiy, S.R., Kuznetsov, V.L. and Ponikaravskih, A.E. (2007) Universal Portable X-Ray Apparatus. Russian Federation Patent No. 2005503167.
 Mozharova I.E. and Kuznetsov V.L. (2011) The Pulse X-Ray Diagnostic Portable Apparatus “Yasen 01”. Medical business, 9, 84-85.