The spatial resolution of the gamma-rays camera was measured on a60Co gamma-rays source with edge method. The gamma-rays camera is consisting with rays-fluorescence convertor, optical imaging system, MCP image intensifier, CCD camera, electronic control system and other devices, and is mainly used in the image diagnostics of the intense pulse radiation sources . Due to the relatively big quantum detective efficiency (DQE) and quantum gain of the gamma-rays, etc., the experimental data were processed by averaging multiple images and fitting curves. According to the experimental results, the spatial resolution MTF (modulation transfer function) at the 10% intensity was about 2lp/mm. Meanwhile, because of the relatively big dispersion effects of the fluorescence transmissions in the scintillator and the optical imaging system, the maximal single-noise ratio (SNR) of the camera was found to be about 5:1. In addition, the spatial resolution of the camera was measured with pulse X-rays with 0.3MeV in average energy and exclusion of the effects of secondary electrons from consideration. Accordingly, the spatial resolution MTF at the 10% intensity was about 5lp/mm. This could be an additional evidence to verify the effects of secondary electrons induced by the 1.25MeV gamma-rays in the scintillator upon the spatial resolution. Based on our analysis, the dispersion sizes of the secondary electrons in the scintillator are about 0.4mm-0.6mm. Comparatively, as indicated by the detailed analysis of the spatial resolutions of the MCP image intensifier and CCD devices, both of them have little effect on the spatial resolution of the gamma-rays camera that could be well neglected.
Cite this paper
H. Xie, J. Zhang, J. Chen, F. Zhang, L. Li, J. Qi and Y. Chu, "Evaluation of Major Factors Affecting Spatial Resolution of Gamma-Rays Camera," Journal of Analytical Sciences, Methods and Instrumentation
, Vol. 3 No. 4, 2013, pp. 227-233. doi: 10.4236/jasmi.2013.34029
 S. Watson, T. Kauppila, et al., “The High-Energy Radiographic Machine Emitting X-Rays Flash Radiographic Camera,” SPIE Proceedings, Vol. 2869, 1997, pp. 920-928.
 G. J. Yates and N. S. P. King, “High-Frame Intensified Fast Optically Shuttered TV Cameras with Selected Imaging Applications,” SPIE Proceedings, Vol. 2273, 1997, pp. 126-149.
 P. B. Gree and T. van Doorn, “Evaluation of an Algorithm for the Assessment of the MTF Using an Edge Method,” Medical Physics, Vol. 27, No. 9, 2000, pp. 2048-2059.
 H. Q. Zhu, “Study on the Image Restoration of Y-Ray Pinhole Imaging System,” Ph.D. Thesis, Tsinghua University, Beijing, 2008.
 P. Z. Takacs and I. Kotov, “PSF and MTF Measurement Methods for Thick CCD Sensor Characterization,” SPIE Proceedings, Vol. 7742, 20120, Article ID: 774207.
 A. Karcher and C. J. Bebek, “Measurement of Lateral Charge Diffusion in Thick Fully Depleted, Back-Illuminated CCDs,” IEEE Transactions on Nuclear Science, Vol. 51, No. 5, 2004, pp. 2231-2237. http://dx.doi.org/10.1109/TNS.2004.834721
 H. Schwarzer and A. Boerner, “Dynamic PSF and MTF Measurement on a 9k TDI CCD,” SPIE Proceedings, Vol. 7106, 2008. http://dx.doi.org/10.1117/12.797055
 H. Q. Zhu and K. L. Wang, “MTF Measurement and Analysis of Microchannel Plate Image Intensifiers,” Acta Photonica Sinica, Vol. 36, No. 11, 2007, pp. 1983-1986.
 J. M. Ma and K. L. Wang, “Research on PSF of Scintillator as g-Ray Fluorescenc Image Converter,” Nuclear Electronics & Detection Technology, Vol. 31, No. 4, 2011, pp. 473-478.
 J. H. Zhu and S. L. Niu, “Energy Deposition of Gamma Rays in LSO Crystal,” High Power Laser and Particle Beams, Vol. 22, No. 6, 2010, pp. 1351-1354.