To get better methods on diagnosis of cardiovascular diseases and health assessment by comparison analysis of four kinds of cardiovascular signals, which are blood pressure signal (BPS), plus wave signal (PWS), electrocardiosignal (ECS) and cardiac sound signal (CSS), a four-channel signals collector is introduced in this article. This collector is composed of microcomputer STM32F103RB with integrated 12-bit analog to digital converters (ADCs) and some integrated chips (IC) such as PGA113, a programmable gain amplifier with internal calibration channels for system-level calibration. Therefore, the data of four kinds of cardiovascular signals can be both collected by this signal collector with a high precision and output to a computer by universal serial bus (USB) interface, which can also offers power for this instrument. In addition, the gain of input signal can be changed by computer software through USB interface under microchip commands. Thus it has many merits, such as the small volume, low cost, high performance and precision, digital-controlled gain and so on. Consequently, it is convenient to acquire the data of four kinds of cardiovascular signals by it, and a waveform of one sample is shown.
 Lamarre-Cliché, M., Cheong, N.N.G. and Larochelle, P. (2011) Comparative assessment of four blood pressure Measurement methods in hypertensives. Canadian Journal of Cardiology, 27, 455-460. doi:10.1016/j.cjca.2011.05.001
 Romagnoli, S., Romano, S.M., Bevilacqua, S., et al. (2011) Dynamic response of liquid-filled catheter systems for measurement of blood pressure: Precision of measurements and reliability of the Pressure Recording Analytical Method with different disposable systems. Journal of Critical Care, 26, 415-422. doi:10.1016/j.jcrc.2010.08.010
 Lan, H., Al-Jumaily, A.M., Loweb, A. and Hing, W. (2011) Effect of tissue mechanical properties on cuff-based blood pressure measurements. Medical Engineering & Physics, 33, 1287-1292. doi:10.1016/j.medengphy.2011.06.006
 Pickering, T.G., et al. (2005) Recommendations for blood pressure measurement in humans and experimental animals: Part 1: Blood pressure measurement in humans: A statement for professionals from the Subcommittee of Professional and Public Education of the American Heart Association Council on high blood pressure research. Hypertension, 45, 142-161.
 STMicroelectronics (2013) Medium-density performance line ARM-based 32-bit MCU with 64 or 128 KB Flash, USB, CAN, 7 timers, 2 ADCs, 9 com Interfaces. http://www.st.com/st-web-ui/static/active/en/resource/technical/document/datasheet/CD00161566.pdf
 Texas Instruments (2013) Zerø-drift programmable gain amplifier with mux. http://www.ti.com/lit/ds/symlink/pga113.pdf